Daily Archives: August 1, 2014
A little over a year ago, I wrote a post on How to build a better insulin pump, based on the Medtronic Revel. Since very little has changed from the Revel to the 530G, those wishes/recommendations still apply. But now that I’ve had some time to play around with Enlite, I’ve got some wishes/recommendations on how that can be improved upon as well. My reason for posting this is simple — if nobody knows, nothing will change; but if someone (or some-two, or some-many) makes noise, it’s more likely to be heard. With the goal of being taken seriously, I’m keeping my recommendations somewhat simple and realistic.
* * *
The new Enlite and Enlite “Serter” are a vast improvement over the predecessor. In nearly every way. That goes without saying.
When I first saw the Sofsensor, I concluded that this must have been designed in-house by Medtronic, and not by Unomedical, the Masters (and manufacturer)-of-all-things Infusion Set related (and whose website appears to be out-of-commission at the moment). Medtronic’s first CGM lacked the characteristics of something being designed by the Masters.
Enter Enlite. It’s better, and it took into account all the feedback received from Sofsensor users. It, too, has been designed in-house. And sadly, it too lacks some of the characteristics of something designed by the Masters.
Let me be clear: I do think the Enlite is a good product, and this in no way is meant to be criticism worthy of driving someone away from using it. My reason for writing it is so that Medtronic – or perhaps another manufacturer – can learn something and make improvements the next time around. My hope is that they’ll take a look at some of the “little things” that can leave a big impression. And maybe, just maybe, the current product can undergo minor improvements while still being the current product.
Allow me to explain.